Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. Draft saved Draft deleted. Baluncore , Sep 2, I do not have access to this particular third-party instrument driver, but, based on looking at your code, what you probably want to do is set the scale of the XY Graph separately. Message 2 of 4. Sep 2, 9. That is what I see:
|Date Added:||23 March 2018|
|File Size:||27.16 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
That is what I see: Can anyone advise me based on what I have provided, or are there still other settings I am missing? The Data VIs- transfer data to or from the instrument. Can you please help me with all these problems, and tell me if they are related?
Message 7 of Most Active Software Boards: If you need the high voltage and currents you could use an external trigger to produce pulses. However, when trying to run the plug-and-play examples, this error occurs “The ID Query failed. It asks you labvirw uncheck the ID query at the initialise vi and if the driver and zgilent are compatible, it should work.
Lastly, you don’t need to typecast the raw data into an I32 then typecast to SGL real.
Search Instrument Driver Network
We can quickly write the code for a trinket if you specify the frequency sweep parameters. As for the driver compatibility you will have to check the manuals to see if they use the same commands.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 9 of If you hit a piezo structure with a hammer, or a single pulse, it will ring at it’s resonant frequency.
The Close VI- terminates the software connection to the instrument and frees system resources.
Message 4 of 4. Most instrument drivers have menu palettes that include the following components. No, create an account now. You can go directly to SGL. Message 1 of I don’t know where you got the driver, but it is reading the binary block transfer all wrong.
Solved: LabVIEW: Identification Query Failed – Discussion Forums – National Instruments
Message 2 of When I click to install it, I cannot find it actually it does not allow me to proceed to Search because it cannot identify the manufacturer. After you call these VIs, the instrument is ready to take measurements or to stimulate a system.
Then the Configure Measurement VI is called to set the instrument up for the desired measurement. It would be extremely useful to show how to modify portions of instrument drivers agilwnt other use.
The alternative to using the NI Instrument Driver Finder is to go directly to the Instrument Driver Network and search for your instrument driver there.
I hope the snaps below will help to explain the problem: Everyone who loves science is here! In this example, serial communication is used. Last edited by a moderator: You may also be using a model that is not officially supported by this driver. lahview
Agilent Signal Generator – IEEE (GPIB) Driver for LabVIEW – National Instruments
I’ll attach snapshots – once I get back to the lab – if someone find it necessary to understand what exactly I am facing. What is your actual problem?
Do you already have an labviww